Philips Respironics DreamStation CPAP Machine has an accessible and user-friendly menu designed to provide alerts and sleep results for CPAP users. Choose from one of the many detailed, setup options to select flex pressure relief, ramp, or OPTIONAL humidification. Download the Respironics’ DreamMapper app onto a tablet or smartphone to access nightly sleep results relayed through Bluetooth wireless technology. DreamMapper is available on Apple devices with iOS 6 or newer and Android products 2.2 or newer (smartphone and tablet not included).
- Heated Tube Humidifier
- Bluetooth Connection to DreamMapper App
- Ramp and SmartRamp
- C-Flex Technology
- LCD On-Screen Display
- Respironics Dry Box Design
- System One Resistance Control
- Detailed ‘Setup Menu’ Options
Tube temperature and humidification settings can be set while the unit is in operation. Use the control dial to make the selections. Also, Adaptive mode can be enabled which will monitor ambient humidity levels and automatically adjust heat settings to provide optimum humidification while helping reduce rainout.
Bluetooth Connection to DreamMapper App
The DreamStation CPAP has a built-in Bluetooth connection which allows the device to pair directly with the PR DreamMapper mobile and web-based program. Enable the Bluetooth feature and download the SleepMapper app onto a smartphone or tablet (not included) to begin reviewing detailed sleep results. Use the free application to begin accessing nightly results through a personalized account on a handheld device. Please Note: The results are only available on one device at a time.
Ramp and SmartRamp
The Ramp feature and SmartRamp feature are both designed to gradually adjust the user to CPAP therapy. Though the Ramp features are similar, there are a few differences between the two options.
Ramp is intended to gently increase CPAP pressure from half of the prescribed pressure to the full pressure within a chosen period of time. To operate the Ramp feature, a minimum CPAP pressure of 4cm H2O is required. Choose a time ranging from 5 minutes to 45 minutes to allow the machine to build the pressure in increments of 0.5cm H2O before reaching the prescribed pressure.
In CPAP mode, SmartRamp uses an Auto-CPAP algorithm, making adjustments to the decreased pressure based on a breath by breath basis. Once SmartRamp reaches the prescribed pressure, the device will begin operating as a CPAP machine at the prescribed pressure. Please Note: During SmartRamp, if the minimum pressure is not reached before Ramp is complete, the pressure will increase by 1cm H2O each minute following the conclusion of Ramp.
C-Flex Technology is a pressure relief feature designed for CPAP machines. A range of settings numbered 1 through 3 are established to create varying levels of pressure reduction upon exhalation. Setting 1 is used to apply the least amount of change, while Setting 3 offers the greatest ease upon exhalation.
LCD On-Screen Display
PR’s DreamStation Series was built with a simplistic and practical LCD screen. While the User Interface is in use, the screen is well lit. As the lights in the room are dimmed, the screen’s sensor is designed expel the light from the screen. Please Note: The control dial is used to view the menu. The screen does not have touch screen capabilities.
Sleep results are posted on the User Interface to display results and goals for sleep therapy.
The first screen accounts for a “Three Night Summary” which is determined after 12pm on the third day of treatment. Refer to the User Interface to view the results as they update each day.
Respironics Dry Box Design
Dry Box technology was created to help reduce the chance of water damage when the OPTIONAL humidifier is used with the DreamStation machine. In the instance that water may escape from the chamber, the Dry Box is designed to trap the water within the walls of the unit to prevent water from traveling throughout the machine and humidifier connection. Additionally, the air outlet port uses a one way valve to prevent water from exiting the water chamber and entering the machine. The air outlet port is located on the top of the water chamber to help minimize total water displacement.
System One Resistance Control
System One Resistance Control was developed by Respironics to help their machines reach the optimal pressure by using a numbering system which allows the CPAP device to recognize the mask. It achieves accuracy and reliability by instructing the device to compensate for the specific mask type. Please Note: PR masks are the only masks designed with a resistance control number, so this machine feature is only compatible with Philips Respironics masks.
Detailed ‘Setup Menu’ Options
Scroll through the User Interface using the control dial. The “My Setup” section allows the user to enable a variety of features found on the DreamStation device.